FFP2 – N95 Aerofresh Respirator filtering face piece respirators (FFR), which are sometimes called disposable respirators, are subject to various regulatory standards around the world. These standards specify certain required physical properties and performance characteristics in order for respirators to claim compliance with the particular standard. During pandemic or emergency situations, health authorities often reference these standards when making respirator recommendations, stating, for example, that certain populations should use an “N95, FFP2, or equivalent” respirator. This document is only intended to help clarify some key similarities between such references, specifically to the following FFR performance standards: • N95 (United States NIOSH-42CFR84) • FFP2 (Europe EN 149-2001) • FFP2 – KN95 Respirator (China GB2626-2006) • P2 (Australia/New Zealand AS/NZA 1716:2012) • Korea 1st class (Korea KMOEL – 2017-64) • DS (Japan JMHLW-Notification 214, 2018) As shown in the following summary table, respirators certified as meeting these standards can be expected to function very similarly to one another, based on the performance requirements stated in the standards and confirmed during conformity testing. One notable comparison point is the flow rates specified by these standards for the inhalation and exhalation resistance tests. Inhalation resistance testing flow rates range from 40 to 160L/min. Exhalation resistance testing flow rates range from 30 to 95 L/min. Some countries require testing to be performed at multiple flow rates, others at only the high or low end of those ranges. Although this appears to suggest that the standards’ requirements for breathing resistance (also called “pressure drop”) differ from each other, it’s important to understand that pressure drop across any filter will naturally be higher at higher flow rates and lower at lower flow rates. Given typical pressure curves for respirator filters, the FFP2 – N95 Aerofresh standards’ various pressure drop requirements are actually quite similar. This chart shows a representative filter pressure drop curve. If one filter is tested at a high flow rate, the pressure drop performance will be relatively high. If that same filter is tested at a low flow rate, the pressure drop performance will be relatively low.
We support the activities of the World Health Organization. Please read on the website all the advices to be always informed about the latest news of COVID-19. Watch also on our News Section the video advices to prevent your contamination.